Microheater. Low Amp, High Temperature. High Power Microheater. GAXP-IgMp. 1350°C-1600°C.
Temperature: Maximum continuous temperature not to exceed 1500C (2732F). Microheater for 1600C (~2900F) for short burst duration. Common use 1400°C (2552°F), Rapid heat product.
Volts: 24 V each.
Size: Heated length is 1″. Total cylinder length is 2.2″ +_0.5″. Lug on each side is 0.5″. Total length ~3.2″
Heated Section Length ~1″
Power: Up to 400W
Element GAXP-IgMp. The more accurate measurement technique at these temperatures is a radiation pyrometer.
Electricals: Maximum power is 400 Watts. The maximum temperature is achieved for ~ 17A with full power and maximum temperature when the S24-400 is used in moderately free radiative conditions (upto 20A RMS meter read-out in fully free-radiating conditions)
A 24V minimum supply rated for 17 Amps (variable) is the minimum required for heating to glowing conditions.
A 24V DC (nominal) power generators/supply is offered. in the pull down menu. This is not a feedback control. For Feed back Control choose BPAN-O-PLUS and Power Transformer that MHI will recommend.
Heating Rate: The heater will start getting hot as soon as power is introduced to it. Typically one will notice a glow (i.e when it is above ~800°C) in the order of a minute, when unit is powered in a normal manner.
Connectors: Accessory Pack M6 Bolts (picture shown) may be added.
Physical Details: Heater length is 1″. Total cylinder length including lugs ~2.2″. Not including the rings or prongs required for the electrical connections. Changes slightly with the terminal contacts chosen. Heated Section Length ~12mm+-1mm (~0.5″).
The heater diameter is ~9 +-0.6 mm (nominal) (Not including the terminal forks/rings shown in the picture). Near the lugs the diameter is ~22mm.
The maximum heater amps (~17 Amps- 20 Amps ) or temperature (which should be measured with a calibrated pyrometer) cannot be exceed. Please call MHI for emissivity settings.