Useful for steam oxidation, testing steam reactions in the laboratory, testing for environmental innovation and food or antimicrobial testing purposes
The chambers may be used for steam up 1300C. Independent control of steam, gas and chamber with industry tested control panels.
Accurate sample positioning and multiple temperature read outs are offered.
Please contact MHI for details to specify the steam flow rate and temperature and the sample size and temperature.
These are one of the best test chambers as they allow for continuous-flowing clean steam in order to provide steady state test conditions with time independent steam-gas.
Steam Output: 1 kg/hr.
HGA Models: Pure Steam
300°C (HGA-S-01) to
Model S-Chamber -01-300 1Kg/hr 300C steam
Model S-chamber -01-1300 1Kg/hr 1300C steam
Steam Rate Flow Adjustment: Discrete and continuous mode is available
Test in chamber with premixed steam air mixtures with HGA-M.
Steam output: 4-12 kg/hr. OAB® Models
Models (4kg/hr, 12kg/hr)
Single phase 208-240V.
Model S-Chamber-04-300 4Kg/hr 300C steam
Model S-Chamber-04-1200 4Kg/hr 1200C steam
Temperature Range: 300-1200°C
Steam Rate Flow Adjustment: Discrete and continuous modes for steam flow adjustments.
Up to 4 Bar Back Pressure Tolerant Models Available.
(Based on questions from previous sales).
(a) Are there recommended minimum & maximum coupon size for superheated sample testing? The chamber 1.5 inch diameter & 4 inch length
(b) What is the incremental distance (upper and lower) between coupon and the steam applicator?
Infinite type Adjustment
(c) Are there any constraints in varying coupon exposure time to superheated steam? (e.g. minimum exposure time & max exposure time?)
(d) Are the coupon (or) steam applicators, both are fixed? In other words, if we need to vary sweeping rate, can this be done? (it may be related to question c).
Sample Holder is Multi-axis # Tilt Positions
Fixed or you can continuously change.
(e) presume we can monitor humidity within the chamber with appropriate probe sensor during the experiments
Please see https://mhi-inc.com/PG2/steam-superheated-humidity-generator.html. Humidity is not defined for high temperature steam
(f) What are the upper and lower limits of temperature for steam that can be used during the experiment? at what increment temperature can be controlled?
Up to ~400C or 1000C or 1300C. Infinite control is allowed by MHI control systems.
(e) can the steam velocity be varied?
Not without changing nozzles. A ¼” nozzle is provided. Note that the volume flow rate is fixed.
(f) Are there any constraints with water source (e.g., DI water or tap water is OK?)
RO water is preferred.
(g) Do you offer a Data Acquisition system
Yes. Please inquire about MHI-DACS.
(h) What if I want a moving product steam tunnel
Please link to steam tunnels.
Request a Quote
Want to find out how MHI can help you meet your steam needs?
Please fill out a quick questionnaire by clicking on the product of interest.
We will help find a good solution to meet your needs.
Micropyretics Heaters International Inc.
750 Redna Terrace Cincinnati OH 45215