Useful for steam oxidation, testing steam reactions in the laboratory, testing for environmental innovation, and food or antimicrobial testing purposes
Depending on the model, the chambers may be used for steam up from low-temperature 1300C. Independent control of steam, gas, and chamber with industry-tested control panels.
There are two standard models. Contact for standard models to 300°C and 1300°C for testing, for example, bacterial inactivation in food or for testing for long times in continuously replenished fresh steam conditions.
Accurate sample positioning and multiple temperature readouts are offered.
Please contact MHI for details to specify the steam flow rate, temperature, sample size, and temperature.
These are one of the best test chambers as they allow continuous-flowing clean steam to provide steady-state test conditions with time-independent steam gas.
The smallest chamber size is 1″ID and a 4″ Long, open chamber.
(a) Are there recommended minimum & maximum coupon sizes for superheated sample testing? The chamber is 1.5-inch in diameter & 4-inch in length.
(b) Can gas be introduced?
Yes. with the gas port attachment.
(c) What is the total distance (upper and lower) between the coupon and the steam applicator? Can I vary the steam rate? Yes, with a CN-BPE or BPE. The CN-BPE, if offered for the model, offers continuous adjustment from 0.3Kg/hr to 1 Kg/hr.
(d) Are there any constraints in varying coupon exposure time to superheated steam? (e.g., minimum exposure time & max exposure time)?
(e) Are the coupon (or) steam applicators both fixed? In other words, can this be done if we need to vary the rate? (it may be related to question c).
Sample Holder is Multi-axis # Tilt Positions
Fixed, or you can continuously change.
(f) presume we can monitor humidity within the chamber with an appropriate probe sensor during the experiments